Question regarding a FLuke A meter display. «on: January 28, , 04 am». So I’ve gotten into restoring “vintage” DMMs from Ebay. I’ve gotten. Download FLUKE A IM.V4 SERVICE MANUAL service manual & repair info for electronics experts. Hi guys. I managed to buy a Fluke A for £1, knowing that it wasn’t working. The selection switches were broken bar one, and they were.
|Published (Last):||27 June 2005|
|PDF File Size:||3.39 Mb|
|ePub File Size:||4.51 Mb|
|Price:||Free* [*Free Regsitration Required]|
Clarence House, 31, Clarence St. At the end of the integrate period Q8 opens and Q4 and Q6 close. Second, the display will not indicate any applied dc input; producing a displayed readout of plus two or three digits of noise.
Compare the current DVM reading with that taken in step 3. Incorrect output voltages from any of the power supplies may cause the multimeter to exhibit various improper indications.
Fluke 8810A Digital Multimeter
New data will start to load into the DOU after the next measurement 88810a complete. I’ve read the manual http: California Stackpole Components Co. The listing of this manual is to help those who need this manual to repair 8810a maintain their equipment. If the voltage continues to rise, Q2 will also be connected in the series and the total standoff will then be 1 kV.
Senen Raya 44, P. Chelmsford, Massachusetts Aumet Corp. High voltage applied to the protection circuit is enough to make appreciable reverse current to flow through Q No ratings or reviews yet. Mansfield, Ohio West Hartford.
Lockwood, Florida Ferroxcube Corp. Jul 16, Manuel Montt – Of.
Q6 may be open or the gate signal may not be present during the read period. With reference to Figureuse the following procedure to install the Rack Mounting Kit.
The following procedure is used to gain access to the interior of the A: Therefore, its trailing edge is recommended for use as a print command to the recorder. Parts lists include the following information: Fasten the rack mount brackets and retainers to the case as shown. Determine the measurement functions that will be available when the True RMS Conver- ter is installed; i.
City of Industry, California Vernitron Corp. The waveform at TP I can indicate several possible problems in the circuitry. Chicago, Illinois Bell Industries Elect. This section contains information regarding the installation and operation of the Model A. Install the top guard cover on the A and perform the Calibration adjustments pro- cedure given later in this subsection.
Fluke 8810A Digital Multimeter Missing Rubber Foot Pad – Used
The maximum voltage limits that may be applied between adjacent input terminals without damaging the A are given in Table As each input signal is applied, adjust the specified adjustment for an 88 10A display reading within the limits shown.
Measurements Restrictions This item may be a floor model or store return that has been used. B C V5C 3H3 Carefully inspect the interior of the instru- ment: Remove link connecting guard terminal and low input. Carlisle, Pennsylvania Bourns Inc. The process effectively takes the square root of the squared input signal and presents it to the input of the integrator amplifier. Disconnect the wires from the connector posts on the Display PCB.
Fluke A Angle Position Indicator w/On- Board Ref.,Progrmable FLV $ For Rent | Transcat
Free Run allows the DOU output to be automatically updated at the end of each measurement cycle. Recommended Handshaking Connections OA negative converter input will cause the opposite logic level output from each pin. Do not remove the top inner guard cover.
The current in this case is dissipated by resistors R35, R36, and R The A front panel display will generally indicate a buffer malfunction in one of three ways. U10 enables the data on W to be serially loaded into the bottom of U4 during the first strobe cycle.
The display must read 1. See section 5 in any Fluke technical manual for ordering instructions. Save on Multimeters Trending price is based on prices over last 90 days.
Fluke A for Sale|Voltmeters|Meters|Test Equipment Center
Measurement capabilities of the A can be increased by the addition of one or more of the avail- able options and accessories listed in Table If your instrument contains one or more of these options they may be calibrated in any sequence following the previous Calibration Adjustments procedure.
The term, input signal conditioner, describes the basic function of the three flike grouped under it. This allows access to the components on the various printed circuit boards.
Component reference designators mentioned in the text identify the component on the simplified circuit drawings and on the schematic diagrams. Accessories include a rack mounting kit and several specialized probes. Specifications for the DOU are presented in Section 1 of this manual.